For camera problem check troubleshooting section or contact firstname.lastname@example.org
Lumo SDK installation
Download Lumo SDK installation package, click on the link below:
(For Pleora SDK - eBUS SDK 126.96.36.19942)
Lumo SDK installation package (Version 2020_519)
For older Pleora SDK (eBUS SDK 4.1.7) please use the link below:
Lumo SDK installation package (Version 2019_443)
Select camera(s) of interests in the list
Finish installation and restart computer
Start Breeze and select camera under Settings -> Hardware
For more information follow the Prediktera Breeze hardware and settings guide:
Test Lumo SDK communication with the camera
Open command prompt (CMD) and go to the folder:
Type “example2-x64.exe” and press Return
Enter device number and press Return
Then you should see 100 frames being read from the camera.
Copy Lumo SDK dll files
Copy all files from the specified folder
C:\Program Files (x86)\Specim\SDKs\SpecSensor\2020_519\bin\x64
Or (for older Pleora SDK users)
C:\Program Files (x86)\Specim\SDKs\SpecSensor\2019_443\bin\x64
To the Breeze folder:
Lumo log files are located under: C:\ProgramData\Specim
Check that all cables are connected and that all devices are working normally.
National Instrument cameras (e.g. SWIR and SWIR3)
Make sure that the latest National Instrument software is installed:
SisuChema / ViaSpec tray
The tray does not move after connecting in Breeze.
Change limit switch setting
The limit switch configuration for the motor is default set to 1 in Breeze and for some motors, this value should be changed to 0
Please try this:
Open the BreezeProperties.xml that is located in Breeze Workspace.
Change this line from:
Save and close BreezeProperties.xml.
Start Breeze and Re-program the motor